The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Agilent A. 20 MHz Function /. Arbitrary Waveform Generator. Service Guide . Publication Number (order as manual set). Agilent A. Function/Arbitrary Waveform Generator .. trigger from the rear- panel Trig In connector, a manual trigger from the key, or a software (bus) trigger .
|Published (Last):||13 September 2018|
|PDF File Size:||3.79 Mb|
|ePub File Size:||15.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
The first error returned is the first error that was stored.
33220A Function / Arbitrary Waveform Generator, 20 MHz
Sending a text message to the display overrides the display state as set by the DISP command. The variation in frequency of the modulated waveform from the carrier frequency is called the frequency deviation. Comments to this Manuals Your Name. kanual
Index amplitude limitations 56 period modulating frequency frequency limitations 56 burst mode modulating source 88, modulation modes allowed front-panel selection 17 modulating waveform 87 pulse period limitations pulse waveform 70 modulation 85 output impedance period, pulse modulation waveshape affect agioent amplitude See chapter 5 for a complete listing of the errors.
For periods above 10 seconds, the minimum pulse width is greater.
With interpolation enabled defaultthe waveform editor makes a straight-line connection between points. Chapter 3 Features and Functions Burst Mode Burst Mode You can configure the function generator to output a waveform with a specified number of cycles, called a burst.
In some cases, the amplitude limits are determined by the output units selected. When not used as an input, the Trig In connector can 3320a configured as an output to enable the A to trigger other instruments at the same time as its internal trigger occurs.
Chapter 3 Features and Functions Remote Interface Configuration Subnet Mask LAN Subnetting allows the network administrator to divide a network into smaller networks to simplify administration and to minimize network traffic.
For example, if you are currently outputting a 20 MHz sine wave and then change to the ramp function, the function generator agileent automatically adjust the output frequency to kHz the upper limit for ramps.
Page 8 In This Book Quick Start Chapter 1 prepares the function generator for use and helps you get familiar with a few of its front-panel features. A query of this register returns a decimal value which corresponds to the binary- weighted sum of all bits set in the register. The subnet mask indicates the portion of the host address to be used to indicate the subnet.
Then use the knob or numeric keypad to enter the desired frequency. Check the list of parameters to verify that you have used a valid parameter type. Each character in the name can be a letter “a” The default offset is 0 volts for all functions.
Page Chapter 5 Error Messages Execution Errors Settings conflict; not able to sweep noise, sweep turned off The function generator cannot generate a sweep using the noise function.
The edge time is then incremented.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Symmetry represents the amount 332220a time per cycle that the ramp wave is rising assuming that the waveform mxnual is not inverted. The following steps show you how to change the amplitude to 50 mVrms.
MIN selects the lowest frequency allowed for the selected function and MAX selects the highest frequency allowed. Output Configuration Chapter 3 Features and Functions Output Configuration Output Configuration This section contains information to help you configure the function generator for outputting waveforms.
Agilent Technologies 33220A User Manual
PM is very similar to FM, but in PM the phase of the modulated waveform is varied by the instantaneous voltage of the modulating waveform. Symmetry Ramp Waves Applies to ramp waves only.
The default is Sine. Chapter 3 Features and Functions Pulse Waveforms Pulse Manua A shown below, a pulse waveform consists of a period, a pulse width, a rising edge, and a falling edge. The default is NORM. The commands used manuap FSK are listed on the next page.
Query the Standard Event Status Register. Chapter 5 Error Messages Execution Errors Settings conflict; triggered burst not available for noise You cannot use the noise function in the triggered burst mode.
In order to prevent overheating, do not block the flow of air into or out of the instrument. In PWM, the pulse width or duty cycle of the carrier waveform is varied according to the modulating waveform. Agilwnt autoranging disabled, the function generator uses the current amplifier and attenuator settings.