The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Service Guide. Publication Number (order as manual set) The Agilent Technologies A is a 20 MHz synthesized function. Buy Keysight Technologies A Function Generator 20MHz GPIB, LAN, USB A. Operating manual, service manual, quick reference guide, IntuiLink.
|Published (Last):||11 April 2006|
|PDF File Size:||1.58 Mb|
|ePub File Size:||9.8 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Start Quick Start One of the first things you nanual want to do with your function generator is to become acquainted with the front panel.
Output Configuration Chapter 3 Features and Functions Output Configuration Output Configuration This section contains information to 332200a you configure the function generator for outputting waveforms. Comments to this Manuals Your Name. Page Command Separators A colon: The default is Sine.
A list of help topics is also available to manuaal you with several front-panel operations. Chapter 3 Features and Functions Pulse Width Modulation PWM Width Deviation The width deviation represents the variation in width in seconds in the modulated waveform from the width of the original pulse waveform.
You can configure the function generator to automatically recall the power-down state when power is restored. Single, External, or Internal Carrier Waveforms: Instrument State Storage The function generator has five storage locations in non-volatile memory to store instrument states. The commands used for PM are listed on the next page. To avoid multiple waveform changes, you can enable the burst mode after you have set up the other burst parameters.
Make sure that you have selected the proper trigger source and verify that the sweep or burst mode is enabled.
Press the Burst Period softkey and then set the 33220aa to 20 ms using the numeric keypad or the knob and cursor keys. Manuall commands to select the function, frequency, amplitude, and offset of the waveform. The variation in phase of the modulated waveform from the carrier waveform is called the phase deviation, which can vary from 0 to degrees.
Page 50 characters. Page 58 Chapter 3 Features and Functions Output Configuration Output Frequency As shown below, the output frequency range depends on the function currently selected.
After the specified number of cycles have been output, the function generator stops and waits for the next trigger. Press the key to view the list of available help topics.
33220A Function / Arbitrary Waveform Generator, 20 MHz
To print the manual completely, please, download it. We have written the exercises in this chapter to prepare the instrument for use and help you get familiar with some of its front-panel operations. The factory default is Chapter 7 Tutorial Tutorial Select an internal or external FSK source. The Standard Event enable register is cleared when: However, only one label for example, Freq or Period is displayed for each softkey at one time.
Agilent Technologies 33220A User Manual
Index affect on amplitude 63 PM 85 load termination 35 PWM 93 offset affect on amplitude 63 tutorial description amplitude limitations 60,local language, help 28 modulation depth AM 77,local operation LAN modulation depth, percent modulation arb waveform limitations 61, logarithmic sweep This is the frequency at which the signal on the front-panel Sync connector goes to a logic low during the sweep. MIN selects the most negative dc offset voltage for the selected function and amplitude. You can also control the phase offset from the front panel or over the remote interface.
Chapter 7 Tutorial Modulation Modulation Modulation is the process of modifying a high-frequency signal called the carrier signal with low-frequency information called the modulating signal.
For example, if you select a period using the PULS: While in the waveform editor, you define the waveform by specifying time and voltage values for each point in the waveform. Press the IP Setup softkey.
Phase-lock Commands option Only The rear-panel 10 MHz In and 10 Kanual Out manul present only if Option is installed allow synchronization between multiple Agilent A function generators see connection diagram below or to an external 10 MHz clock signal.
The default is OFF.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Query the Standard Event Status Register. Press the Hop Freq softkey and then set the value to Hz using the numeric keypad or the knob and cursor keys.
Page Index softkey labels 4 questionable data register trigger output software bus trigger,register diagram trigger source standard event register sweeps software revision status byte register Sync signal software, connectivity 15 status msnual symmetry 65 sound beeper